miniaturized pH electrode
The Unisense pH electrode offers you a fast and reliable research tool. With tip sizes down to 10 µm, you can complete high resolution and non-destructive microprofiles in a wide range of microenvironments.
The pH microelectrode is designed for research applications within:
- Environmental sciences
- Biomedical sciences
- Food science e.g. cheese production
This microelectrode is a miniaturized conventional pH electrode based on selective diffusion of protons through pH glass, and the determination of potentials between the internal electrolyte and a reference electrode. The reference electrode can be either external (separate electrode) or internal; however, we only recommend internal references in applications where it is not possible to place two electrodes in the sample.
The Unisense pH microelectrode is supported by the pH/mV meter. This compact and economic one-channel amplifier has excellent signal handling and amplification. The meter also comes with Unisense SensorTrace Logger Software.
part numbers and outside tip diameter
- pH-10: 10-19 µm
- pH-25: 20-30 µm
- pH-50: 40-60 µm
- pH-100: 90-110 µm
- pH-200: 180-220 µm
- pH-200C: 180-220 µm (combination electrode with internal reference electrode)
- pH-500: 400-600 µm
- pH-500C: 400-600 µm (combination electrode with internal reference electrode)
- pH-MR: 400-600 µm
- pH-N: 1.1 mm
- pH-NP: 1.6 x 40 mm – needle for piercing
Reference electrode for pH microelectrodes
Reference electrodes from Unisense are simple silver/silver-chloride open-ended electrodes designed to work with potentiometric microelectrodes (e.g. Unisense pH and Redox microelectrodes). For measurements with these sensors, it is important that the measuring sensor is in fluid contact with the reference. For most purposes this can be achieved with a large macro-reference. For some applications the sample is so small (e.g. a droplet) that the reference must be as small as the measuring microelectrode.
For laboratory use, Unisense provides a macro reference electrode from Radiometer Analytical. Our micro-sized reference electrodes are glass electrodes manufactured at Unisense. For in situ use, Unisense manufactures a pressure-compensated macro electrode.
- Photosynthetic biofilms
- Biogeochemistry and ecosystem functioning
- In-vitro cell culture profiling
- What is biofilm?
Portillo,M.C. et al (2010), Moonmilk Deposits Originate from Specific Bacterial Communities in Altamira Cave (Spain), Microbial ecology, 1 – 8, vol.
Liniger,M. et al (2003), Cleavage of trypanosome surface glycoproteins by alkaline trypsin-like enzyme (s) in the midgut of Glossina morsitans, International journal for parasitology, 1319 – 1328, vol. 33
Kamp,A. et al (2006), Anaerobic sulfide oxidation with nitrate by a freshwater Beggiatoa enrichment culture, Applied and Environmental microbiology, 4755 – 4760, vol. 72